A Krohn-Hite MV106J DC Voltage Standard calibrator landed on Dave’s bench. What’s inside? How does it work? Is it still within spec?, and what’s all this (lack of) Kelvin-Varley and LM299 zener reference stuff anyhow?
And no adventure in calibration check is complete without a visit to the TrioSmartCal lab.
Dave compares the new Siglent SSA3021A 2.1GHz spectrum analyser with similar priced Rigol DSA815. Noise ...