A Krohn-Hite MV106J DC Voltage Standard calibrator landed on Dave’s bench. What’s inside? How does it work? Is it still within spec?, and what’s all this (lack of) Kelvin-Varley and LM299 zener reference stuff anyhow?
And no adventure in calibration check is complete without a visit to the TrioSmartCal lab.
Dave looks at some issues with fanning out tiny 0.4mm pitch BGA packages, via pad ...