EEVblog #433 – MailbagPosted on March 2nd, 2013 3 comments
There is no design intention to the behaviour of the Pulse Generator after the power supply gives up, I just don’t care. ^_^
Logic “1″ & “0″ mode allows me to measure the logic level with a DMM, as well as measuring the output impedance using compensated 4-wires resistance measurement.
Dave Dave Dave… The Motorola battery pack do state the type, it clearly says Nickel Metal Hydride Battery on it. Just under the Motorola logo.
Leave a reply