A follow-up on another method to calibration adjust the Agilent 53131A frequency counter against the Stanford Research PRS10 rubidium standard.
The oscillator drift video is HERE
Forum Topic HERE
A look at the Caesium beam time and frequency standards at the Keysight standards lab ...
The EEVblog Store generally ships twice a week, on Tuesdays & Fridays, Sydney time. Dismiss