A follow-up on another method to calibration adjust the Agilent 53131A frequency counter against the Stanford Research PRS10 rubidium standard.
The oscillator drift video is HERE
Forum Topic HERE
Repairing the Data IO universal programmer. Will it be a happy ending? https://theamphour.com/the-amp-hour-99-impavid-ideopraxist-insider/ Forum HERE ...
The EEVblog Store generally ships twice a week, on Tuesdays & Fridays, Sydney time. Dismiss