Well today I got the meter from UK , I do not know If I can be called as unlucky , or lucky ,
because one confused story happened .
By unpacking the meter , the display was injured ... I can not tell if I had fall as victim in a scam .
All that I can tell was that I had freak out ... was the only damage that I was not expect to see.
The meter was the model 001 , with the rechargeable batteries in it,
the condition of them was " empty " .
The meter was not power up at all , I open it up , found the batteries , and I started to charging them with external PSU .. By miracle ... as soon the two NICAD batteries got some charge ,
by connecting them back on the meter , the meter came to life !!
And the charger of it , started to charge them by it self !! its 2X1.2 Volts 2.200mA medium size batteries.
Charging voltage 2.8 Volts
The meter as soon it got alive , it tested by the Fluke 87-5 ,
and it works accurately in all the DC volt scales.
Its an Fluke made in Holland , all fuses inside was original , and no signs of any preview repair .
I tested continuity , and amperes , everything looks ok .
I notified the seller about the display damage , and I got refunded on the spot , getting back about 19 EUR of the total of 23 EUR , and lots of apologies from the seller .
Now that I have bring to life the meter , I need to find one good display for it,
even if the one that it has , are partially working .
I truly need some help here , I need some one from US, to locate for me one healthy display,
used or new , and we will sort out the details , by PM or email ..
Here is some pictures of the meter damaged , of the meter functional
and the Fluke display it self .
I found the LCD display pictures , from an old sale on ebay.