NOTE: This DS2072 display data was Manually Created to test the DSO measuring system.
so I can better understand .my first DSO.
THIS IS NOT A REAL SIGNAL
The display shows that the measurements are calculated/measured on the data that is displayed.
NOTE that the Sin(x)/x interpolation is applied to the sample data points before measurements as shown by the "Vpp = 7.12V" , the peaks of the Sin ripples ( the data is a step of only 6.0V)
NOTE that the system knows only 2GSa/s is used , so cannot measure under 0.5ns thus rise time is measured as "rise<500ps"
NOTE the difference among Max, Top, and Amp measurements.
This maybe helpful to others to understand this DSO