Part 1 HERE
Dave follows a more methodical troubleshooting procedure in order to track down the fault in the Tektronix TDS3054 oscilloscope.
Measuring the low and high frequency differential outputs of the hybrid front end, the voltage rails and other parts. And has the magic smoke escaped?
And he couldn’t resist playing with his new Flir E8 thermal imaging camera .
Reverse engineering the Micsig DP10007 high voltage differential probe. Turning the PCB into a schematic. ...