Part 1 HERE
Dave follows a more methodical troubleshooting procedure in order to track down the fault in the Tektronix TDS3054 oscilloscope.
Measuring the low and high frequency differential outputs of the hybrid front end, the voltage rails and other parts. And has the magic smoke escaped?
And he couldn’t resist playing with his new Flir E8 thermal imaging camera .
Teardown of the new $389 12bit Rigol DHO800 oscilloscope! Forum: https://www.eevblog.com/forum/blog/eevblog-1563-new-$389-12bit-rigol-dho800-scope-teardown/ EXCLUSIVE Unboxing and first ...