Why do digital oscilloscopes appear noisier than traditional analog oscilloscopes? PART 2
Learn how and why your digital storage oscilloscope display the noise that it does.
In this 2nd installment Dave shows how waveform update rate affects the apparent displayed noise of a waveform, and alters the waveform update rate on an Agilent DSOX3000 to show this.
Also, how sample rate and timebase settings affects the boxcar averaging high-resolution display function.
PART 1 is HERE
Gav “The Mechatronics Guy” explains circular polarised light and demonstrates his physical laser cut models ...