0 Members and 1 Guest are viewing this topic.

Keithley Model 2400 - detected, S/N 0788393, Version: C32 Oct 4 2010 14:20:11/A02 /H/HPre-cal K2400 Calibration data0.2 DCV Range, SENS:DATA? = -1.375303E+00,-1.721084E-05,-1.375407E+00,+0.000000E+000.2 DCV Range, SOUR:DATA? = +2.764157E+05,+9.152109E+02,-2.764087E+05,+9.467695E+022.0 DCV Range, SENS:DATA? = -1.375422E+00,-1.811981E-05,-1.375288E+00,+0.000000E+002.0 DCV Range, SOUR:DATA? = +2.761708E+04,+2.052578E+02,-2.761636E+04,+2.083945E+0220.0 DCV Range, SENS:DATA? = -1.375513E+01,-1.375198E-03,-1.375385E+01,+0.000000E+0020.0 DCV Range, SOUR:DATA? = +2.761523E+03,+2.085977E+02,-2.761447E+03,+2.050625E+02200.0 DCV Range, SENS:DATA? = -1.375421E+02,-1.382446E-02,-1.375290E+02,+0.000000E+00200.0 DCV Range, SOUR:DATA? = +2.761715E+02,+2.086133E+02,-2.761635E+02,+2.050547E+021.0e-06 DCI Range, SENS:DATA? = -6.845910E-07,-9.356427E-11,-6.845222E-07,+0.000000E+001.0e-06 DCI Range, SOUR:DATA? = +5.548882E+10,+2.040664E+02,-5.548042E+10,+2.104687E+021.0e-05 DCI Range, SENS:DATA? = -6.890184E-06,-5.320544E-10,-6.889600E-06,+0.000000E+001.0e-05 DCI Range, SOUR:DATA? = +5.513167E+09,+2.019297E+02,-5.512313E+09,+2.127148E+021.0e-04 DCI Range, SENS:DATA? = -6.869998E-05,-4.991307E-09,-6.869353E-05,+0.000000E+001.0e-04 DCI Range, SOUR:DATA? = +5.529379E+08,+2.017656E+02,-5.528538E+08,+2.128203E+021.0e-03 DCI Range, SENS:DATA? = -6.857473E-04,-4.994217E-08,-6.856770E-04,+0.000000E+001.0e-03 DCI Range, SOUR:DATA? = +5.539511E+07,+2.017305E+02,-5.538684E+07,+2.128320E+021.0e-02 DCI Range, SENS:DATA? = -6.872036E-03,-5.178154E-07,-6.871379E-03,+0.000000E+001.0e-02 DCI Range, SOUR:DATA? = +5.527761E+06,+2.018398E+02,-5.526910E+06,+2.127383E+021.0e-01 DCI Range, SENS:DATA? = -6.876468E-02,-5.178154E-06,-6.875831E-02,+0.000000E+001.0e-01 DCI Range, SOUR:DATA? = +5.524166E+05,+2.018516E+02,-5.523274E+05,+2.127734E+021.0e+00 DCI Range, SENS:DATA? = -6.887156E+00,-5.303621E-04,-6.886814E+00,+0.000000E+001.0e+00 DCI Range, SOUR:DATA? = +5.518421E+04,+8.602578E+02,-5.517694E+04,+9.663477E+02HP 3458A detected TEMP? = 37.5 CDCV Calibration step -0.2 mV , measured : -1.999948835E-01 VDC, deviation -25.583 ppmDCV Calibration step 0.2 mV , measured : 1.999991793E-01 VDC, deviation -4.104 ppmDCV Calibration step -2 V , measured : -1.999991983E+00 VDC, deviation -4.008 ppmDCV Calibration step 2 V , measured : 1.999986310E+00 VDC, deviation -6.845 ppmDCV Calibration step -20 V , measured : -1.999999159E+01 VDC, deviation -0.421 ppmDCV Calibration step 20 V , measured : 1.999993647E+01 VDC, deviation -3.177 ppmDCV Calibration step -200 V , measured : -1.999974777E+02 VDC, deviation -12.612 ppmDCV Calibration step 200 V , measured : 1.999964677E+02 VDC, deviation -17.662 ppmDCI Calibration step -1 uA , measured : -9.999818293E-07 ADC, deviation -18.171 ppmDCI Calibration step 1 uA , measured : 9.999825290E-07 ADC, deviation -17.471 ppmDCI Calibration step -10 uA , measured : -9.999831268E-06 ADC, deviation -16.873 ppmDCI Calibration step 10 uA , measured : 9.999901602E-06 ADC, deviation -9.840 ppmDCI Calibration step -100 uA , measured : -9.999936766E-05 ADC, deviation -6.323 ppmDCI Calibration step 100 uA , measured : 9.999918902E-05 ADC, deviation -8.110 ppmDCI Calibration step -1 mA , measured : -9.999886353E-04 ADC, deviation -11.365 ppmDCI Calibration step 1 mA , measured : 9.999875093E-04 ADC, deviation -12.491 ppmDCI Calibration step -10 mA , measured : -9.999853206E-03 ADC, deviation -14.679 ppmDCI Calibration step 10 mA , measured : 9.999953624E-03 ADC, deviation -4.638 ppmDCI Calibration step -100 mA , measured : -9.999813773E-02 ADC, deviation -18.623 ppmDCI Calibration step 100 mA , measured : 9.999842788E-02 ADC, deviation -15.721 ppmDCI Calibration step -1 A , measured : -9.999645533E-01 ADC, deviation -35.447 ppmDCI Calibration step 1 A , measured : 9.999774736E-01 ADC, deviation -22.526 ppmDCI Verification step Zero A , measured : -1.788568192E-11 ADCDCI Verification step -1 nA , measured : -9.993935778E-10 ADC, deviation -606.422 ppmDCI Verification step 1 nA , measured : 9.888950700E-10 ADC, deviation -11104.929 ppmDCI Verification step -10 nA , measured : -1.000216363E-08 ADC, deviation 216.363 ppmDCI Verification step 10 nA , measured : 9.988446699E-09 ADC, deviation -1155.330 ppmDCI Verification step -100 nA , measured : -1.000165300E-07 ADC, deviation 165.300 ppmDCI Verification step 100 nA , measured : 1.000043848E-07 ADC, deviation 43.848 ppmDCI Verification step -1 uA , measured : -9.999055619E-07 ADC, deviation -94.438 ppmDCI Verification step 1 uA , measured : 9.998880573E-07 ADC, deviation -111.943 ppmDCI Verification step -10 uA , measured : -9.999861188E-06 ADC, deviation -13.881 ppmDCI Verification step 10 uA , measured : 9.999910553E-06 ADC, deviation -8.945 ppmDCI Verification step -100 uA , measured : -9.999887831E-05 ADC, deviation -11.217 ppmDCI Verification step 100 uA , measured : 9.999839386E-05 ADC, deviation -16.061 ppmDCI Verification step -1 mA , measured : -9.999803701E-04 ADC, deviation -19.630 ppmDCI Verification step 1 mA , measured : 9.999745446E-04 ADC, deviation -25.455 ppmDCI Verification step -4 mA , measured : -4.000350359E-03 ADC, deviation 87.590 ppmDCI Verification step 4 mA , measured : 4.000267999E-03 ADC, deviation 67.000 ppmDCI Verification step -10 mA , measured : -9.999839193E-03 ADC, deviation -16.081 ppmDCI Verification step 10 mA , measured : 9.999906841E-03 ADC, deviation -9.316 ppmDCI Verification step -20 mA , measured : -2.000352007E-02 ADC, deviation 176.003 ppmDCI Verification step 20 mA , measured : 2.000418054E-02 ADC, deviation 209.027 ppmDCI Verification step -100 mA , measured : -1.000000364E-01 ADC, deviation 0.364 ppmDCI Verification step 100 mA , measured : 9.999876289E-02 ADC, deviation -12.371 ppmDCI Verification step -500 mA , measured : -5.001218448E-01 ADC, deviation 243.690 ppmDCI Verification step 500 mA , measured : 5.001075087E-01 ADC, deviation 215.017 ppmDCI Verification step -1 A , measured : 1.000053606E+00 ADC, deviation 53.606 ppmDCI Verification step 1 A , measured : -1.000009712E+00 ADC, deviation 9.712 ppmDCI Verification step Zero A , measured : 1.384233768E-08 ADCDCV Verification step 00 (0.000000000E+00), measured : -8.871845474E-06 VDCDCV Verification step 01 (-2.000000000E-03), measured : -2.010135899E-03 VDC, deviation 5067.950 ppmDCV Verification step 02 (2.000000000E-03), measured : 1.990735405E-03 VDC, deviation -4632.297 ppmDCV Verification step 03 (-2.000000000E-02), measured : -2.001099592E-02 VDC, deviation 549.796 ppmDCV Verification step 04 (2.000000000E-02), measured : 1.999615615E-02 VDC, deviation -192.193 ppmDCV Verification step 05 (-2.000000000E-01), measured : -2.000073866E-01 VDC, deviation 36.933 ppmDCV Verification step 06 (2.000000000E-01), measured : 1.999953503E-01 VDC, deviation -23.248 ppmDCV Verification step 07 (-1.000000000E+00), measured : -1.000055863E+00 VDC, deviation 55.863 ppmDCV Verification step 08 (1.000000000E+00), measured : 1.000063591E+00 VDC, deviation 63.591 ppmDCV Verification step 09 (-2.000000000E+00), measured : -2.000200776E+00 VDC, deviation 100.388 ppmDCV Verification step 10 (2.000000000E+00), measured : 2.000241744E+00 VDC, deviation 120.872 ppmDCV Verification step 11 (-1.000000000E+01), measured : -1.000062483E+01 VDC, deviation 62.483 ppmDCV Verification step 12 (1.000000000E+01), measured : 1.000049089E+01 VDC, deviation 49.089 ppmDCV Verification step 13 (-1.500000000E+01), measured : -1.500109513E+01 VDC, deviation 73.009 ppmDCV Verification step 14 (1.500000000E+01), measured : 1.500119825E+01 VDC, deviation 79.883 ppmDCV Verification step 15 (-2.000000000E+01), measured : -2.000004998E+01 VDC, deviation 2.499 ppmDCV Verification step 16 (2.000000000E+01), measured : 2.000003855E+01 VDC, deviation 1.928 ppmDCV Verification step 17 (-1.000000000E+02), measured : -1.000070850E+02 VDC, deviation 70.850 ppmDCV Verification step 18 (1.000000000E+02), measured : 1.000052509E+02 VDC, deviation 52.509 ppmDCV Verification step 19 (-2.000000000E+02), measured : -1.999974277E+02 VDC, deviation -12.861 ppmDCV Verification step 20 (2.000000000E+02), measured : 1.999966999E+02 VDC, deviation -16.500 ppmDCV Verification step 21 (0.000000000E+00), measured : -2.192978188E-04 VDCPost-cal K2400 Calibration data0.2 DCV Range, SENS:DATA? = -1.375304E+00,-1.619756E-05,-1.375398E+00,+0.000000E+000.2 DCV Range, SOUR:DATA? = +2.764154E+05,+9.151484E+02,-2.764104E+05,+9.473320E+022.0 DCV Range, SENS:DATA? = -1.375421E+00,-1.704693E-05,-1.375296E+00,+0.000000E+002.0 DCV Range, SOUR:DATA? = +2.761707E+04,+2.052461E+02,-2.761638E+04,+2.084570E+0220.0 DCV Range, SENS:DATA? = -1.375512E+01,-1.420975E-03,-1.375377E+01,+0.000000E+0020.0 DCV Range, SOUR:DATA? = +2.761523E+03,+2.086992E+02,-2.761452E+03,+2.049922E+02200.0 DCV Range, SENS:DATA? = -1.375419E+02,-1.417542E-02,-1.375297E+02,+0.000000E+00200.0 DCV Range, SOUR:DATA? = +2.761712E+02,+2.087305E+02,-2.761636E+02,+2.049766E+021.0e-06 DCI Range, SENS:DATA? = -6.845868E-07,-9.220003E-11,-6.845208E-07,+0.000000E+001.0e-06 DCI Range, SOUR:DATA? = +5.548899E+10,+2.039805E+02,-5.548046E+10,+2.105508E+021.0e-05 DCI Range, SENS:DATA? = -6.890186E-06,-5.265974E-10,-6.889579E-06,+0.000000E+001.0e-05 DCI Range, SOUR:DATA? = +5.513168E+09,+2.018594E+02,-5.512315E+09,+2.127813E+021.0e-04 DCI Range, SENS:DATA? = -6.870023E-05,-4.889444E-09,-6.869372E-05,+0.000000E+001.0e-04 DCI Range, SOUR:DATA? = +5.529382E+08,+2.016289E+02,-5.528539E+08,+2.129570E+021.0e-03 DCI Range, SENS:DATA? = -6.857464E-04,-4.877802E-08,-6.856792E-04,+0.000000E+001.0e-03 DCI Range, SOUR:DATA? = +5.539507E+07,+2.016250E+02,-5.538670E+07,+2.129336E+021.0e-02 DCI Range, SENS:DATA? = -6.872036E-03,-5.168840E-07,-6.871358E-03,+0.000000E+001.0e-02 DCI Range, SOUR:DATA? = +5.527748E+06,+2.017773E+02,-5.526900E+06,+2.128164E+021.0e-01 DCI Range, SENS:DATA? = -6.876391E-02,-5.215406E-06,-6.875829E-02,+0.000000E+001.0e-01 DCI Range, SOUR:DATA? = +5.524202E+05,+2.018477E+02,-5.523324E+05,+2.127852E+021.0e+00 DCI Range, SENS:DATA? = -6.887750E+00,-5.336404E-04,-6.886620E+00,+0.000000E+001.0e+00 DCI Range, SOUR:DATA? = +5.518539E+04,+8.598516E+02,-5.517846E+04,+9.664961E+02Calibration complete!