EEVblog #659 – Medical Plugpack TeardownPosted on September 2nd, 2014 6 comments
What’s inside an IEC60601-1 medical class 5V mains DC-DC plugpack?
What is the difference between PCB clearance and creepage?
A German FRIWO 5V FW7555M/05 brand medical grade isolation unit vs a cheap Wun Hung Lo consumer brand.
Also a comparative look at transformer construction.