Dave answers a beginner forum question on why the Rigol DS1054Z oscilloscope does not show the intended vertical Volts/Div setting he expects.
A trap for young and old players alike.
1X and 10X probes, and auto probe interfaces.
The forum question is HERE
Other related videos:
EEVblog #601 – Why Digital Oscilloscopes Appear Noisy
EEVblog #279 – How NOT To Blow Up Your Oscilloscope!
EEVblog #453 – Mysteries of x1 Oscilloscope Probes Revealed
EEVblog #396 – Bode Plotting on Your Osciloscope